EMFC MK III E.L.F. Magnetic Field Compensation
To maximize performance:
- Make sure detector is aligned with the axes of the room (x,y,z)
Disable all channels. Close door of BOTH EM rooms. Turn off video
monitors. Enable all channels.
Plug BNC cable into X channel and observe on oscilloscope. Use small
screwdriver to increase gain of X channel until feeback observed on
oscilloscope. Reduce gain by 0.5 turns.
- Repeat same procedure for Y channel and Z channel.
- Do this sequence several times until system converges.